BODY (Outer Diameter)
TIP (Outer Diameter)
1/8" flow cell
5 to 15 seconds
DEPTH of IMMERSION:
10 uS - 100 mS
-5°C to 75° C
TYPE: Solution & Electrode
Pumping a known calibration standard (0.01 KCl) will help in determining the K for the electrode
Using any electrode in solutions containing protein requires the electrode be rinsed with an enzyme cleaning solution.
After each use, we recommend cleaning our electrodes with Terg-a-zyme (Alconox, Inc.) or a chromic/sulfuric acid glass cleaning solution by submerging the electrodes for a couple of minutes in order to remove all protein from the glass and reference junction.
This will prolong the useful life of the electrodes.
Always clean the microelectrode before storing:
- Long-term (over 2 weeks): rinse flow path with DI water and dry
- Short-term: Rince flow path with DI water