BODY (Outer Diameter)
TIP (Outer Diameter)
2.4 4 mm
5 to 15 seconds
DEPTH of IMMERSION:
50 uS - 100 mS
-5°C to 100° C
Use a known standard (0.01 KCl) to determine the K of the electrode
Using any electrode in solutions containing protein requires the electrode be rinsed with an enzyme cleaning solution.
After each use, we recommend cleaning our electrodes with Terg-a-zyme (Alconox, Inc.) by submerging the electrodes for a couple of minutes in order to remove all protein from the glass and reference junction.
This will prolong the useful life of the electrodes.
Always clean the microelectrode before storing:
- Long-term (over 2 weeks): Rinse with DI water and dry.
- Short-term: Rinse with DI water.